Mass standards and weights are used to verify the metrological characteristics of balances, scales and mass comparators. Specified mass, measurement uncertainty and traceability feature in mass standards, while weights have suitable accuracy class. Section 3.7 of EN 45501 'Non-automatic weighing instruments' standard dictates that the requirements of OIML R 111-1 recommendation must be fulfilled by the mass standards. Section 5.2 specifies the essential requirements. Expanded uncertainty:
For each weight, the expanded uncertainty, U, for k = 2, of the conventional mass, no more than one-third of the maximum permissible error in Table 1 .
The readability of the tested object needs to be taken into account first when selecting weights for testing. The dimensions, shape, material, surface smoothness and maximum permissible error all define the weight, due to legal requirements. The maximum permissible error of weight, if the accuracy class is known, is less than the value given in table 1, OIML R111-1. In the case of very accurate measurements, the possibility of using such a weight is complicated when its conventional mass is not known.
Calibration is a determination of a weights real mass and uncertainty. Currently, when the weight is subjected to calibration, it becomes a mass standard. Therefore, two measuring instruments can be used simultaneously if they feature the same design.
Figure 1. Set of weights from 1 mg to 200 g. Image Credit: Radwag Balances & Scales
Figure 2. Weight and mass standard basic differences. Image Credit: Radwag Balances & Scales
To define, carry out, maintain or reproduce a unit of mass, measuring equipment called mass standards is used. These can be of any shape and any material, which ensures the stability of mass over time. An identification and calibration certificate must be available for mass standards providing details on traceability and estimated measurement uncertainty. Measurement uncertainty estimated during the calibration process forms the basis of the elementary classification for mass standards. Mass standards cannot be used as weights within the meaning of legal metrology.
The units of measure are weights, determined by the following documents: OIML R111-1 and ASTM E617. Previously, weighing instruments of I and II accuracy classes were also adjusted using weights. Metrological requirements for weights, in terms of compulsory verification worldwide, have been specified by the International Legal Metrology Organisation. OIML R111 (2004) refers to weights ranging from 1 mg to 50 kg. The requirements are specified in terms of accuracy classes, material, shape, identification and protection.
A hierarchical proportion 1:3 is used to determine the accuracy classes E1 up to M3, where E1 is the highest class and M3 is the lowest. Supervision over measuring equipment has to be carried out as periodic tests and/or calibration with the use of measuring equipment, mass standards and weights (section 126.96.36.199. of ISO 9001). The results of such tests have to be documented.
 Bettin H., Physikalisch-Technische Bundesanstalt (PTB), S Schlamminger, National Institute of Standards and Technology (NIST), “Realization, maintenance and dissemination of the kilogram in the revised SI”, Metrologia 53 (2016) A1–A5.
 Chung J., Ryu K., Davis R., “Uncertainty analysis of the BIPM susceptometer”, Metrologia 38 (2001), pp. 535-541.
 Davis R., “New method to measure magnetic susceptibility”, Meas. Sci. Technol. 4, 141–147 (1993).
 Davis R., “Determining the magnetic properties of 1 kg mass standards”, J. Res. Natl. Inst. Stand. Technol. 100, 209 (1995).
 Guidelines on the Calibration of Non-Automatic Weighing Instruments EURAMET/cg-18/v.02, January 2009.
 Janas S., “Metrology in laboratory – measurement mass and derived values” Radwag, 2015.
 Janas S., “Micro scale measurements”, Radwag 2014.
 Fujii K., Bettin H., Becker P., Massa E., Rienitz O., Pramann A., Nicolaus A., Kuramoto N., Busch I. , Borys M., “Realization of the kilogram by the XRCD method”, Metrologia 53 (2016) A19–A45.
 Lu H., Chang C., “Evaluating the magnetic property of one kilogram mass standard in Center for Measurement”, APMF'2003 Proceedings, pp. 57-60.
 Mohr P. J., Newell D. B. and Taylor B. N., 2016 CODATA recommended values of the fundamental physical constants: 2014 Rev. Mod. Phys. (arXiv:1507.07956).
 Myklebust T. 1997, “Intercomparison: Measurement of the volume magnetic susceptibility and magnetization of two cylindrical (kg) weights. EUROMET project 324”, Justervesenet (NO).
 Myklebust T., Davis R., “Comparison between JV and BIPM to determine the volume susceptibility of one 20 g weight and two 1 g weights”, Justervesenet (1996).
 Myklebust T., Källgren H., Lau P., Nielsen L., Riski K., “Testing of weights: Part 3 - Magnetism and convection”, OIML Bulletin, Vol. XXXVIII, No. 4 (1997) 5.
 OIML D 28, “Conventional value of the result of weighing in air” Edition 2004 (E).
 OIML R 111-1, “Weights of classes E1, E2, F1, F2, M1, M1–2, M2, M2–3 and M3, Part 1: Metrological and technical requirements”.
 Pan Sheau-shi, Lu H., Chang C., “Instruments used for measuring the magnetic properties of one kilogram mass standard in Center for Measurement Standards (CMS)”, TC3 IMEKO’2005 Proceedings, pp. 1-7.
 Robinson I., Schlamminger S., “The Watt or Kibble balance: a technique for implementing the new SI definition of the unit of mass”, Metrologia 53 (2016) A46–A74.
 Richard P., Fang H., Davis R., “Foundation for the redefinition of the kilogram”, Metrologia 53 (2016) A6–A11.
 Stock M., “The Watt balance: determination of the Planck constant and redefinition of the kilogram”, Bureau International des Poids et Mesures (BIPM), Pavillon de Breteuil, Downloaded from http://rsta.royalsocietypublishing.org/ on March 13, 2017.
About Radwag Balances & Scales
Our mission is designing professional products based on technically advanced concepts.
The ideas are realized by supplying the customers with more and more innovative products, which are employed by most demanding end users with great satisfaction. Measurable effects of our activity are positive opinions of numerous users, thanks to which RADWAG is placed among leading manufacturers of balances and scales.
Sponsored Content Policy: News-Medical.net publishes articles and related content that may be derived from sources where we have existing commercial relationships, provided such content adds value to the core editorial ethos of News-Medical.Net which is to educate and inform site visitors interested in medical research, science, medical devices and treatments.